|Place of Origin:||China|
|Minimum Order Quantity:||1 set|
|Packaging Details:||wooden box|
|Payment Terms:||L/C, T/T, Western Union|
|Power Supply:||AC220V ± 10%, 50Hz||The Heating Device:||Quartz Tube Furnace Heating|
|Designated Sweeping Device:||Automatic Sweep Program.||, Thermometer::||(-6 ~ 400) ℃|
|Ambient Temperature:||-10 ℃ ~50 ℃.||Power Consumption:||Less Than 650W|
SYD-3536 Cleveland Open Flash Point Tester
The main technical characteristics and parameters
1 Power supply: AC220V ± 10%, 50Hz.
2, the heating device: quartz tube furnace heating, no fire, explosion, power 0~600W continuously adjustable.
3, designated sweeping device: automatic sweep program.
4, thermometer: (-6 ~ 400) ℃, dividing 2 ℃, technical conditions are met by petroleum products specified liquid thermometer technical conditions GB / T514's.
5, ignition device: ignition sources: gas (or other civilian combustible gas, the same below);
spout aperture: (0.6~0.8) mm.
6, ambient temperature: -10 ℃ ~50 ℃.
7, relative humidity: ≤85%.
8 Power consumption: less than 650W
The main structure
1, ignition sweep stroke switch: press this switch, turn the ignition igniter forward; press the switch again, turn the ignition igniter reverse.
2, ignition: ignition assembly for connecting the gas source and the point of the gas at the top nut is used to adjust the flame size.
3, ammeter: electric heating is used to display current, regulation and control of heating power.
4, electric stove: electric heating.
5, Cleveland Oil Cup: placing a sample of the instrument and a dedicated heating oil cup.
6, the thermometer frame: a fixed thermometer and keep it vertical.
7, the regulator knob: electric power for heating regulation.
8. Power Switch: Turn this switch, indicator light, the instrument it is powered.
1. Warranty 1 year of replacement and return.
2, Providing factory visit and study, technical training
3, 24-hour technical support by e-mail or phone
4, Providing laboratory construction program design guidelines
5, Providing product design and OEM service