|Place of Origin:||China|
|Model Number:||SYD - 3536 - D|
|Minimum Order Quantity:||1 set|
|Packaging Details:||wooden box|
|Payment Terms:||L/C, T/T, Western Union|
|Supply Ability:||100 sets/month|
|Temperature Measurement:||Range: Room Temperature ~ 400 ℃;||The Ambient Temperature:||(10 ~ 40) ℃;|
|Total Power Consumption::||Less Than 500W||Heating Rate:||In Line With GB / T3536 Standard|
YD-3536D Automatic opening flash point tester
It is based on the People's Republic of China standard GB / T 3536 "Petroleum products - Determination of flash and fire (Cleveland open cup method)" requirement set forth in the design and manufacture, according to the standards applicable to prescribed method, determination of fuel oil and opening flash point below 79 ℃ petroleum products Cleveland open cup flash point Lanfu removed.
The instrument color LCD screen, using SCM control technology, all Chinese human-machine interface, unmarked keyboard; flash point temperature value to be pre-sample label, atmospheric pressure, test dates and other parameters, with menu prompts guiding input function; application open, fuzzy control integration software, modular design techniques.
The major technical indicators and parameters
1 Power supply: AC220V ± 10%, 50Hz;
2, temperature measurement: Range: room temperature ~ 400 ℃;
Repeatability: ≤4 ℃;
Reproducibility: ≤8 ℃;
Temperature display unit: 0.1 ℃;
3, heating rate: in line with GB / T3536 standard;
4. Ignition: electronic ignition, the gas flame 3mm ~ 4 mm;
5, the ambient temperature: (10 ~ 40) ℃;
6, relative humidity: ≤80%;
7 Total power consumption: less than 500W.
The instrument structure
Power switch: installed on the left side of the instrument, turning the switch on, the instrument is powered on.
printer: Mini dot matrix printer, the test ends automatically print test results.
Display: color LCD screen, real-time display test results and other information.
operation key: the instrument without marking key operation, according to the screen prompts to operate.
draw Kon: The fireball lit the timing required to draw sweeping glass surface of the sample.
ignition: is a vertical cross fold up the structure, is the instrument of the electronic ignition.
fire mouth: as an upright structure, as required during operation is always lit, as designated Kon ignition sources.
boom: the instrument boom can be lifted when the sample cups were placed in the sample cup holder, put the instrument into the test mode.
Note: rise boom decline is done automatically, must not forcibly pull on Banxia hand to avoid damaging boom!
detection ring: an annular structure, whether the test sample flash point, ignition detection sensor.
Note: The detection loop should be kept clean, otherwise it will reduce the detection sensitivity!
temperature sensor: mounted on the lift arm, specimen cup, temperature detection sample is placed in the test. specimen cup holder: Place the sample cup.
cover: usually placed on the right inside the instrument, when the instrument detects and after 5 seconds after ignition, automatic unscrew the cover plate specimen cup, to prevent accidents.
flame adjustment knob: used to adjust the fuse head, draw Kon size of the flame (gas supply source, the size of the output valve via air conditioning).
outlet: located on the back of the instrument, the configuration of the power cord plugs in here.
air intakes: in the back of the instrument, through which the gas supply source to the instrument.
1. Warranty 1 year of replacement and return.
2, Providing factory visit and study, technical training
3, 24-hour technical support by e-mail or phone
4, Providing laboratory construction program design guidelines
5, Providing product design and OEM service
Packing and Shipping: